Time of flight diffraction and imaging (TOFDI)
Petcher, P. A. (2011) Time of flight diffraction and imaging (TOFDI). PhD thesis, University of Warwick.
WRAP_THESIS_Petcher_2011.pdf - Submitted Version
Download (11Mb) | Preview
Official URL: http://webcat.warwick.ac.uk:80/record=b2581406~S1
Time of flight diffraction and imaging (TOFDI) is based on time of flight diffraction (TOFD),
adding cross-sectional imaging of the sample bulk by exploiting the scattering of ultrasonic waves
from bulk defects in metals. Multiple wave modes are emitted by a pulsed laser ultrasound ablative
source, and received by a sparse array of receiving electromagnetic acoustic transducers (EMATs),
for non-contact (linear) scanning, with mode-conversions whenever waves are scattered.
Standard signal processing techniques, such as band-pass filters, reduce noise. A B-scan is
formed from multiple data captures (A-scans), with time and scan position axes, and colour
representing amplitude or magnitude. B-scans may contain horizontal lines from surface waves
propagating directly from emitter to receiver, or via a back-wall, and angled lines after reflection
off a surface edge. A Hough transform (HT), modified to deal with the constraints of a B-scan,
can remove such lines. A parabola matched filter has been developed that identifies the features
in the B-scan caused by scattering from point-like defects, reducing them to peaks and minimising
noise. Multiple B-scans are combined to reduce noise further. The B-scan is also processed to form
a cross-sectional image, enabling detection and positioning of multiple defects.
The standard phase correlation technique applied to camera images, has been used to track
the relative position between transducer and sample. Movement has been determined to sub-pixel
precision, with a median accuracy of 0.01mm of linear movement (0.06 of a pixel), despite uneven
illumination and the use of a basic low resolution camera.
The prototype application is testing rough steel products formed by continuous casting, but
the techniques created to facilitate operation of TOFDI are applicable elsewhere.
|Item Type:||Thesis or Dissertation (PhD)|
|Subjects:||Q Science > QC Physics|
|Library of Congress Subject Headings (LCSH):||Time measurements, Diffraction, Metals -- Surfaces -- Defects, Metals -- Nondestructive testing|
|Official Date:||September 2011|
|Institution:||University of Warwick|
|Theses Department:||Department of Physics|
|Supervisor(s)/Advisor:||Dixon, Steven Mark, 1969-|
|Sponsors:||University of Warwick ; Engineering and Physical Sciences Research Council (EPSRC) ; Corus Group|
|Extent:||xiv, 179 leaves : ill., charts|
Actions (login required)
Downloads per month over past year